The authors describe the construction of an electronic model for teaching the supraclavicular and interscalene approaches to brachial plexus block. Realistic feedback is provided to indicate contact with the plexus or pleura. All components are readily available, and cost of construction should be less than $150 provided a skeleton is available to provide a frame for the model.
- Brachial plexus
- Regional anesthesia
- Resident training
Statistics from Altmetric.com
Presented as a scientific exhibit at the annual meeting of the American Society of Anesthesiologists, Las Vegas, Nevada, 1982.
Address reprint requests to Dr. Gross: Department of Anesthesia, Philadelphia Veterans Administration Medical Center, University and Woodland Avenues, Philadelphia, PA 19104.
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.