Background Many epidural and peripheral nerve catheters contain conducting wire that could heat during magnetic resonance imaging (MRI), requiring removal for scanning.
Methods We tested 2 each of 6 brands of regional analgesia catheters (from Arrow International [Reading, Pennsylvania], B. Braun Medical Inc [Bethlehem, Pennsylvania], and Smiths Medical/Portex [Keene, New Hampshire]) for exposure to clinical 1.5- and 3-T MRI. Catheters testing as nonmagnetic were placed in an epidural configuration in a standard human torso–sized phantom, and an MRI pulse sequence applied at the maximum scanner-allowed radiofrequency specific absorption rate (SAR) for 15 minutes. Temperature and SAR exposure were sampled during MRI using multiple fiberoptic temperature sensors.
Results Two catheters (the Arrow StimuCath Peripheral Nerve and B. Braun Medical Perifix FX Epidural) were found to be magnetic and not tested further. At 3 T, exposure of the remaining 3 epidural and 1 peripheral nerve catheter to the scanner’s maximum RF exposure elicited anomalous heating of 4°C to 7°C in 2 Arrow Epidural (MultiPort and Flex-Tip Plus) catheters at the entry points. Temperature increases for the other catheters at 3 T, and all catheters at 1.5 T were 1.4°C or less. When normalized to the body-average US Food and Drug Administration guideline SAR of 4 W/kg, maximum projected temperature increases were 0.1°C to 2.5°C at 1.5 T and 0.7°C to 2.7°C at 3 T, except for the Arrow MultiPort Flex-Tip Plus catheter at 3 T whose increase was 14°C.
Conclusions Most but not all catheters can be left in place during 1.5-T MRI scans. Heating of less than 3°C during MRI for most catheters is not expected to be injurious. While heating was lower at 1.5 T versus 3 T, performance differences between products underscore the need for safety testing before performing MRI.
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.
This work was supported in part by National Institutes of Health grant R01 EB007829 (P.A.B.).
The authors declare no conflict of interest.